What are the steps to treat a dental implant screw fracture?
Dental implant screw fracture on you and need a solution? A separation or fracture of the abutment screw is not a good day. In my experience most are loose and come out easily with a piezo. If it doesn’t move immediately with the piezo I have never gotten them out without out drilling them out. The dental code for a broken screw removal is D6096.
When the dental implant screw is loose
If the screw is loose there are many tricks you can use.
- Place a small piece or rubber band in the access, cotton, or maybe even plumbers tape to give a better grip
- Sharp explorer
Steps for dental implant screw fracture removal with piezo
- Tap with piezo gently and see if spinning.
- If it is spinning will usually come out pretty easily fairly quickly.
- If not moving at all may be cold welded and not going to come out at all.
Steps for dental implant screw fracture removal with fragment fork
- Clean area and you better have high power loupes, 4.5x or higher, and a headlight. Check to see if thread are damaged (if damaged check steps below)
- Place correct size Fragment Fork in slow speed and center on fragment
- Apply force to ensure engaged
- Reverse torque
Steps for dental implant screw fracture without anything special
- Endo explorer, make sure it is sharp, turn the broken portion of the screw clock-wise 1/2 to 1 full turn, then counter-clockwise the same amount you turned in the screw piece in the first place.
- Repeat this several times.
- Do not screw in more than 1 full turn!!!
- This will release the bur on the screw that occurred when the screw broke. It is tedious.
Mechanical kit without guide tube may be the fastest Panadero 2017
- 1/4 round bur in handpiece place a dimple in center screw
- Place a buccal and lingual groove from center dimple
- Form a screwdriver out of a #1 bur by cutting it with a stone.
- Use fork to insert screw fragment deeper
- Use retapping tool (Zimmer has one) Yilmaz JPD Aug 2015
- Use fork again to try to remove now
- Use ultrasonics as needed
How to check for internal damage after screw removal?
After a screw is out there may be internal damage if one of the methods you use can cut metal. To check this you can simply inject impression material into the implant and place a wedge of some sort into the PVS before it sets. When you remove the impression you can visually check for any obvious damage.
Other prosthetic problems
Implant with damaged internal threads
Companies sell tapping systems that retread the internal aspects but I wonder how well that really works. Another option is to simply make a post and cement the post into a prepared internal aspect. The picture below was an even more complicated case but the idea is the same.
Fracture zirconia abutment
Dentaltown thread on when a zirconia abutment breaks
JPD study from 2017 from Saponaro showing the novel use of a plastic perio probe to rock the piece back and forth.